Hiv Test Kit Rapid 3rd Generation
HIV Test kit
for the diagnosis of Human Immunodeficiency Virus ( HIV ) to detect antibodies to HIV type 1, type 2 and Subtype O qualitatively in whole blood Serum or plasma
199 in stock (can be backordered)
Hiv Test Kit Card
2 Hiv Test Kit trial pack
HIV Test Kit is Highly sensitive even to HIV IgG IgM during early HIV infection stage of 1-4 weeks
Differentiated test result between HIV type I and II by clear band formation(3-lines)
Specimen : Serum, Plasma (10uL), or Whole blood(20uL)
Sensitivity : 100%
Specificity : 99.8%
2 HIV Test Kit
hiv test kit
A rapid test for the diagnosis of Human Immunodeficiency Virus to detect antibodies to HIV type 1, type 2 and Subtype O qualitatively in whole blood Serum or plasma. For professional in vitro diagnostic use only. The HIV 1/2/O Rapid Test Cassette (Whole Blood/Serum/Plasma) is a rapid chromatographic immunoassay for the qualitative detection of antibodies to Human Immunodeficiency Virus(HIV) type 1,type 2 and subtype O in whole blood, serum or plasma to aid in the diagnosis of HIV infection. HIV (Human Immunodeficiency Virus) is the etiologic agent of Acquired Immune Deficiency Syndrome (AIDS). The virion is surrounded by a lipid envelope that is derived from the host cell membrane. Several viral glycoproteins are on the envelope. Each virus contains two copies of positive-sense genomic RNAs. HIV-1 has been isolated from patients with AIDS and AIDS-related 1 complex, and from healthy people with high potential risk for developing AIDS. HIV-1 consists of Subtype M and Subtype O. Highly divergent strains of HIV-1 were first recognized in 1990 and grouped provisionally as Subtype O as this variation has similar glycoprotein markers to HIV-1 but a slight variation to the protein marker. Although rarely compared to HIV-1 and HIV-2, infections caused by Subtype O have so far been identified in Africa (Cameroon), France and Germany. HIV-2 has been isolated from West African AIDS patients and from seropositive asymptomatic 2 3 individuals. HIV-1, HIV-2, and Subtype O all elicit immune responses. Detection of HIV antibodies in serum, plasma or whole blood is the most efficient and common way to determine 4 whether an individual has been exposed to HIV and to screen blood and blood products for HIV. Despite the differences in their biological characters, serological activities and genome sequences, 5,6 HIV-1, HIV-2, and Subtype O show strong antigenic cross-reactivity. Most HIV-2 positive sera can be identified by using HIV-1 based serological tests. The HIV 1/2/O Rapid Test Cassette (Whole Blood/Serum/Plasma) is a rapid test to qualitatively detect the presence of antibodies to HIV type 1, type 2, and/or Subtype O in whole blood, serum or plasma specimen. The HIV 1/2/O Rapid Test Cassette(Whole Blood/Serum/Plasma) is a qualitative, membrane based immunoassay for the detection of antibodies to HIV-1, HIV-2, and Subtype O in whole blood, serum or plasma. The membrane is pre-coated with recombinant HIV antigens in the test line regions, T1 and T2. The T1 test line is pre-coated with HIV-1 and Subtype O antigen and the T2 test line is precoated with HIV-2 antigen. During testing, the whole blood, serum or plasma specimen reacts with HIV antigen coated particles in the test strip. The mixture then migrates upward on the membrane chromatographically by capillary action and reacts with recombinant HIV antigen on the membrane in the test line region. If the specimen contains antibodies to HIV-1 and/or Subtype O, or HIV-2, one coloured line will appear in the test line region; if the specimen contains antibodies to HIV-1 and/or Subtype O, and HIV-2, two coloured lines will appear in the test line region. Both indicate a positive result. If the specimen does not contain HIV-1, Subtype O, and/or HIV-2 antibodies, no coloured line will appear in the test line region indicating a negative result. To serve as a procedural control, a coloured line will always appear in the control line region indicating that proper volume of specimen has been added and membrane wicking has occurred. The test contains HIV type 1, type 2, and Subtype O recombinant antigens coated particles and HIV type 1, type 2, and Subtype O recombinant antigens coated on the membrane. • For professional in vitro diagnostic use only. Do not use after expiration date. • Do not eat, drink or smoke in the area where the specimens or test cassettes are handled. • Do not use test if pouch is damaged. • Handle all specimens as if they contain infectious agents. Observe established precautions against microbiological hazards throughout all procedures and follow the standard procedures for proper disposal of specimens. • Wear protective clothing such as laboratory coats, disposable gloves and eye protection when specimens are assayed. • The used test should be discarded according to local regulations. • Humidity and temperature can adversely affect results. Store as packaged in the sealed pouch either at room temperature or refrigerated (2-30°C). The test is stable through the expiration date printed on the sealed pouch. The Hiv test kit must remain in the sealed pouch until use. DO NOTFREEZE. Do not use after the expiration date. The HIV 1/2/O Rapid Test Cassette (Whole Blood/Serum/Plasma) can be performed using whole blood (from venipuncture or fingerstick), serum or plasma. Testing should be performed immediately after the specimens have been collected. Do not leave the specimens at room temperature for prolonged periods. Serum and plasma specimens may be stored at 2-8°C for up to 3 days. For long term storage, specimens should be kept below -20°C. Whole blood collected by venipuncture should be stored at 2-8°C if the test is to be run within 2 days of collection. Do not freeze whole blood specimens. Whole blood collected by fingerstick should be tested immediately. • Bring specimens to room temperature prior to testing. Frozen specimens must be completely thawed and mixed well prior to testing. Specimens should not be frozen and thawed repeatedly.
hiv test kit POSITIVE:* Two or three distinct coloured lines appear. One line should always appear in the control line region (C), and another one or two apparent coloured line(s) should appear in the test line region(s) (T1 and/or T2). *NOTE: The intensity of the colour in the test line region (T1 and T2) will vary depending on the concentration of HIV antibodies present in the specimen. Therefore, any shade of colour in the test line region (T1 and/or T2) should be considered positive. hiv test kit NEGATIVE: One coloured line appears in the control region (C). No apparent coloured lines appear in the test line regions (T1 and T2). INVALID: Control line fails to appear. Insufficient specimen volume or incorrect procedural techniques are the most likely reasons for control line failure.
Legal Disclaimer: this is an diagnostic product to be used in or by medical professional expert supervision only Review the procedure and repeat the test with a new test device. If the problem persists, discontinue using the test kit immediately and cross verify with ELISA / PCR Methods at your diagnostic centre or referral diagnostic laboratory
Note: This Product Not Available on Cash On Delivery ( COD Orders will be Cancelled )
Name: Microsidd India
Address: 3048/14, Mkk Road, Gayatrinagar,
Google Location: https://g.page/r/CWXyv3vvzsGPEAE
Only logged in customers who have purchased this product may leave a review.
- 5.00 rating from 2 reviews